Due to green algae problems in the water reservoir of a drinking water treatment plant, the ALGcontrol was installed to continuously monitor the algae concentrations and possible toxic blue green algae blooms. During almost 4 months the ALGcontrol measured the total chlorophyll and cyano chlorophyll concentrations of the water reservoir. In order to be able to compare these results, second off-line controls were carried out using the ISO 10260 chlorophyll-a quantification (extraction into ethanol, spectrophotometric detection at 665 and 750 nm). The results of the off-line standard lab method show a good correlation with the measurements of the ALGcontrol. The good correlation can be verified as well at high as at low concentrations of total chlorophyll.
By microLAN B.V. based in Waalwijk, NETHERLANDS.
Need help finding the right suppliers? Try XPRT Sourcing. Let the XPRTs do the work for you