Sound Instrumentation for Environmental Noise Monitoring - Health and Safety - Noise and Vibration
State of the art noise monitoring systems; Modern noise monitoring systems offer a wide range of advanced features and come with software that can analyze and structure long term measurements data efficiently. A complete noise monitoring system consists of one or more noise monitoring terminals, a noise server and work stations for presentation, post processing and reporting.
Noise Monitoring Terminal
The noise monitoring terminal is the piece of equipment that acquires sound level and other data and transmits it to the server. It comprises the sound level measurement apparatus (microphone, power supply, sound level meter) along with an enclosure, a communication device and possibly a weather station and camera.
The bases for a noise monitoring station are the microphone and the sound level measurement device. In order to be certain that your measurements are of the highest quality, the sound level meter should be approved to satisfy the standard IEC 61672:2003 'Electroacoustics - sound level meters' by a national institution such as PTB (the German national metrology institute). An approved meter is tested to give correct results under varying physical conditions. The standard defines the precision of two categories of meters: Class 1 and Class 2. A Class 1 meter is required for noise measurements in most countries.
Microphones for permanent outdoor installations must be designed to withstand variations in temperature and humidity, and also resist heavy rain and corrosive climate. They should have a self-check functionality using an electrostatic actuator or a charge injection method. This way the microphones can be automatically verified so that the interval for manual inspection of the terminals can be prolonged. Still, the microphones are subject to a yearly calibration in an accredited laboratory to ensure lasting accuracy. Regular laboratory calibration is also required if measurements taken shall be valid for court cases
Additionally a weather station might be required to do proper analysis of the measurements; especially important is wind direction and wind speed. With advanced software for noise analysis all measurements done during high wind speed causing a high noise level measured by the microphone can be easily extracted from the measurement results.
The noise monitoring terminal should be robustly designed with battery backup, tamper proof enclosure and proper temperature regulation to secure stable long term logging of noise data. It should also have a self-monitoring function so that the status of the terminal (such as external power and battery status, door openings etc) is logged and can be verified by the operator.
The communication revolution with broadband data traffic even wirelessly has opened for a new world of noise monitoring. Some years ago the most advanced systems would only allow transmission of small amounts of data. Now, transmission capacity has increased and communication prices have decreased. This means that higher resolution results can be transmitted and even sound recordings and video images may be streamed close to real time. State of the art noise monitoring terminals support all modern communication platforms such as ADSL, WIFI and 3G mobile networks. An acoustic consultant can therefore have access to noise measurements from a number of monitoring stations in her office and measurements can be made available real time on the Internet.