This paper describes a simple approach to minimizing probe-induced flow distortion errors in a three-axis sonic anemometer. By separating the three axes, mutual interference between the transducers and supports in the three arrays is reduced. Only a transducer shadow correction determined from wind tunnel tests is needed to obtain measurements that are virtually insensitive to probe orientation relative to the mean flow. Preliminary results from a series of three atmospheric tests verify the effectiveness of this correction.
- Applied Technologies, Inc. (ATI)
- Minimizing flow distortion errors in a sonic anemometer
Case study - Barnsley Council
As part of their commitment to improving the energy efficiency of all homes and buildings within their borough, Barnsley Council contacted Logic Energy to see if they could assist Dearne Carrfield Primary School in discovering their “green potential” before investing in a renewable energy system. With solar thermal systems, wind, solar PV and biomass systems all an option, the school wanted a monitoring system that would measure various meteorological sensors to help them to decide on the most...
What are the advantages of ultrasonic wind sensors?
Whether mechanical, ultrasonic or laser wind measurements: All measuring methods have technical limits. While for mechanical wind sensors the measuring element – cups and vanes – itself predominantly produces a disturbance in the flow field, for ultrasonic anemometers a disturbance can be caused by the sensor holder or housing construction. Only optical methods, e.g. LDA (Laser Doppler Anemometer), enable a disturbance-free measurement of wind speeds. However, for this particles are required and such...
Introducing The RNRG Goalpost Boom
We are happy to introduce a new goalpost mounting solution for installing redundant class 1 anemometers. This new boom, designed specifically for use with our 80m XHD TallTower, positions anemometer cups at exactly 80 m above ground level—2.5 m away from each other—and 1.99 m above the horizontal arm of the boom. This configuration complies with guidelines provided in Annex G of the IEC 61400-12-1:2005 standard when installed on a RNRG 80m XHD TallTower—a solution that is well-suited for conductin...
Luminant Oak Grove Power Station - Case Study
Powering Texas Luminant, a subsidiary of Energy Future Holdings Corp. (EFH) -which owns competitive and regulated energy subsidiaries in Texas -is the largest electricity generator in Texas. Unit two of the Oak Grove Power plant near Franklin in Robertson County, TX went online in the middle of 2010. Fluor, the project’s EPC company, completed the plant in 35 months and used advanced control technologies to make the Oak Grove super-critical generating station one of the cleanest coal-fired power plants in...
Metek Introduces the New Multi-Path Ultrasonic Anemometers
Our news report about recent instrumentations or measurements of our customers which might be of special interest for other web site visitors. The reports are added on the fly whenever we have received such information. The experienced support is highly welcome and appreciated. Metek is proud to present the new members of the uSonic-family: The ultrasonic anemometers uSonic-3 Class A MP and uSonic-3 Cage MP represent an innovative step forward to highest performance in atmospheric turbulence sounding. Based on...