Tech Tip: Deploying Sea-Bird CTD in Ice
Large numbers of Sea-Bird conductivity instruments have been used in Arctic and Antarctic programs. Special accommodation to keep temperature, conductivity, oxygen, and optical sensors at or above 0° C is advised. Often, the CTD is brought inside protective doors between casts to achieve this.
When freezing is possible, we recommend that the conductivity sensor be stored dry. Remove larger droplets of water by blowing through the cell. Do not use compressed air, which typically contains oil vapor. Attach a length of Tygon tubing to each end of the conductivity cell to close the cell ends. See Application Note 2D: Instructions for Care and Cleaning of Conductivity Cells for details.
In general, neither the accuracy of the temperature measurement nor the survival of the temperature sensor will be affected by ice.
For the SBE 43 and SBE 63 Dissolved Oxygen sensors, avoid prolonged exposure to freezing temperature, including during shipment. Do not store with water (fresh or seawater), Triton solution, alcohol, or glycol in the plenum. The best precaution is to keep the sensor indoors or in some shelter out of the cold weather.