A tracer gas technique has been developed by the Norwegian Institute for Air Research (NILU) to study emission rates, dilution and diffusion of gases. The technique has been applied in several countries to quantify diffuse VOC leakages from refineries, petrochemical plants and oil production platforms.
The tracer gas technique The dual tracer gas system developed by NILU has been applied in several field studies in Europe. Easy-to-use portable release systems, syringe-based samplers and portable gas chromatographs are developed for various applications. Several field studies linked to model development have been reported. The system has been designed to meet the following requirements:
easy to modify,
easy to handle,
samplers with preset timers,
rapid analyses in field.
The release system
The release system can be modified for different scales of application to give concentrations of from 10 ppt to 105 ppt (parts per trillion). Normally the tracer gases are released from 10 litres cylinders containing the liquified gas. Through a reduction valve and a flowmeter the gas release can be kept constant at the desired rate. The two tracers (SF6 and CBrF3) can be analysed simultaniously by the portable field gas chromatographs are.
The air samples are collected in inexpensive syringes integrated over short time intervals (grab samples) or over a preset time period. Normally the time period is 15 minutes. A number of battery-operated samplers have been developed with a timer system for presetting of start/stop. These samplers are small, light and handy. One person can easily carry 24 samplers in field.
Tracer sample analyses
The samples can be analysed in the field by portable gas chromatographs immediately after the collection. The technique thus allows the results of one experiment to be used in the design of the next.
Fast response mobile SF6 detector
Also semi instantaneous sequential samplers and a continous sampler have been developed by NILU. The latter instrument provide instantaneous tracer concentrations in ambient air with an accuracy of <10 ppt. It can be used as a `mobile plume detector` during the set-up of cross wind concentration measurements. The instrument can also be connected to a logging unit to provide time series of concentration fluctuations at fixed locations in the plume. Typical logging frequency would be around 1 Hz.
Tracer gas data evaluation and presentation software has been developed at NILU. This enables tracer gas concentration mapping on a gridded coordinate system specified for the area studied. It also plots cross-wind concentration profiles and estimates cross-wind integrated concentrations. These results are used in the evaluation of sample representativeness, and support the further evaluation of emissions and emission estimates.
The NILU tracer technique has been used in many different applications, e.g. to study:
transport and dispersion,
transport in complex terrain,
traffic tunnel ventilation rates,
ventilation in street canyons,
building and room ventilation rates,
industrial emission rates and
diffusive leakages of air pollutants.
Diffusive VOC leakages
One of the more interesting examples is the estimate of diffusive leakages of hydrocarbons (VOC) from petrochemical factory complexes. SF6 was released in different identified release areas at various heights above the surface in the factory area. The release rates were adjusted to match the rough emission assumptions based upon screening studies undertaken prior to the tracer gas experiments.
During near stationary wind conditions and with no significant atmospheric reactions or deposition of hydrocarbons (VOCs) or other release gases between the leakage points and the sampling points, the emission rates of spesific hydrocarbons can be estimated from simple flux considerations.
A model for such estimates has been developed by NILU. The tracer gas technique thus provides a flexible tool for estimating emission rates and evaluating the downwind impact of these emissions on health and the environment.
Fine chemicals are used as active ingredients or additives in the pharmaceutical and food & beverages industries, as well as in agriculture. The cracking of raw materials such as light virgin naphtha or liquefied gas results in intermediate products (ethylene, hydrogen, methane, CO2). At the end of the process chain, these can provide highly pure materials such as niacin (vitamin B3) or pre-stages of vitamins A and E. Process analyzers play an important role in the efficient further processing of cracker...
LNG is natural gas in its liquid state with high energy density, which makes it useful for storage and transportation over long distances from the gas fields to the consumers. LNG is recovered from natural gas in large-scale liquefaction plants. Process automation is a key issue in LNG plants to ensure cost efficient plant operation and high product quality. Process analyzer deliver important data therefore.Siemens, a leader in process analytical instrumentation, has proven over decades its capability to plan,...
In this field study, the impacts of influent loadings and drying-wetting cycles on N2O emission in a subsurface wastewater infiltration (SWI) system were investigated. N2O emitted under different operation conditions were quantified using static chamber and gas chromatograph techniques. N2O conversion rate decreased from 6.6 ± 0.1% to 2.7 ± 0.1% with an increase in hydraulic loading (HL) from 0.08 to 0.24 m3/m2·d. By contrast, N2O conversion rate increased with increasing pollutant loading (PL) up to 8.2 ± 0.5%...
The Axetris LGDs were operated continuously for two years. After intensive laboratory testing, they went out into the field in 2015. Axetris was able to test the functionality of its sensors under real-life conditions thanks to the support of the Linth waste incineration plant.
The gas concentration measuring devices called laser gas detectors (LGD sensors) are used for measuring hazardous substances. When a gas is conducted through the sensor, the respective gas molecules weaken the laser beam. Gas concentratio...
In the realm of natural gas processing, enterprises don’t always find it economical to send products back to the lab and wait for test results. Being able to perform on-the-spot, highly accurate compositional analysis with gas chromatography equipment can reduce operational costs and help companies determine on-the-spot accurate gas levels in a natural gas stream. Here are some of the reasons why more firms are using a portable gas spectrometer in natural gas analyzing for faster, more accurate results.