A tracer gas technique has been developed by the Norwegian Institute for Air Research (NILU) to study emission rates, dilution and diffusion of gases. The technique has been applied in several countries to quantify diffuse VOC leakages from refineries, petrochemical plants and oil production platforms.
The tracer gas technique The dual tracer gas system developed by NILU has been applied in several field studies in Europe. Easy-to-use portable release systems, syringe-based samplers and portable gas chromatographs are developed for various applications. Several field studies linked to model development have been reported. The system has been designed to meet the following requirements:
easy to modify,
easy to handle,
samplers with preset timers,
rapid analyses in field.
The release system
The release system can be modified for different scales of application to give concentrations of from 10 ppt to 105 ppt (parts per trillion). Normally the tracer gases are released from 10 litres cylinders containing the liquified gas. Through a reduction valve and a flowmeter the gas release can be kept constant at the desired rate. The two tracers (SF6 and CBrF3) can be analysed simultaniously by the portable field gas chromatographs are.
The air samples are collected in inexpensive syringes integrated over short time intervals (grab samples) or over a preset time period. Normally the time period is 15 minutes. A number of battery-operated samplers have been developed with a timer system for presetting of start/stop. These samplers are small, light and handy. One person can easily carry 24 samplers in field.
Tracer sample analyses
The samples can be analysed in the field by portable gas chromatographs immediately after the collection. The technique thus allows the results of one experiment to be used in the design of the next.
Fast response mobile SF6 detector
Also semi instantaneous sequential samplers and a continous sampler have been developed by NILU. The latter instrument provide instantaneous tracer concentrations in ambient air with an accuracy of <10 ppt. It can be used as a `mobile plume detector` during the set-up of cross wind concentration measurements. The instrument can also be connected to a logging unit to provide time series of concentration fluctuations at fixed locations in the plume. Typical logging frequency would be around 1 Hz.
Tracer gas data evaluation and presentation software has been developed at NILU. This enables tracer gas concentration mapping on a gridded coordinate system specified for the area studied. It also plots cross-wind concentration profiles and estimates cross-wind integrated concentrations. These results are used in the evaluation of sample representativeness, and support the further evaluation of emissions and emission estimates.
The NILU tracer technique has been used in many different applications, e.g. to study:
transport and dispersion,
transport in complex terrain,
traffic tunnel ventilation rates,
ventilation in street canyons,
building and room ventilation rates,
industrial emission rates and
diffusive leakages of air pollutants.
Diffusive VOC leakages
One of the more interesting examples is the estimate of diffusive leakages of hydrocarbons (VOC) from petrochemical factory complexes. SF6 was released in different identified release areas at various heights above the surface in the factory area. The release rates were adjusted to match the rough emission assumptions based upon screening studies undertaken prior to the tracer gas experiments.
During near stationary wind conditions and with no significant atmospheric reactions or deposition of hydrocarbons (VOCs) or other release gases between the leakage points and the sampling points, the emission rates of spesific hydrocarbons can be estimated from simple flux considerations.
A model for such estimates has been developed by NILU. The tracer gas technique thus provides a flexible tool for estimating emission rates and evaluating the downwind impact of these emissions on health and the environment.
Location: Ipsden, Oxfordshire (UK)Capacity: 960 kWelInput materials: Pig slurry, farm by-products, energy cropsFeatures: Production of 700 Nm³ biomethane, 360 kWel CHP, Supply natural gas to 2,150 households, Fermentation residue as fertiliser38 million kilowatt hours per yearIcknield Gas Ltd. is the first biogas plant in Great Britain with a gas processing facility from EnviTec Biogas AG. Since the end of 2014, the 370 cubic metre plant, along with a 360 kWel combined heat and power plant in Ipsden, Oxfordshi...
Here the original text from G&I:„The Axetris AG LGD F200 self-contained laser gas detector module is designed for integration by original equipment manufacturers (OEMs) of selective catalytic reduction (SCR) systems and monitoring equipment and is based on proprietary technology- enhanced tuneable diode laser spectrometry. The OEM module can measure concentrations from well under 1 ppm to 1000 ppm. This unit combines high sensitivity/selectivity, contactless hot wet measurement with low-cost of ownership,...
Fine chemicals are used as active ingredients or additives in the pharmaceutical and food & beverages industries, as well as in agriculture. The cracking of raw materials such as light virgin naphtha or liquefied gas results in intermediate products (ethylene, hydrogen, methane, CO2). At the end of the process chain, these can provide highly pure materials such as niacin (vitamin B3) or pre-stages of vitamins A and E. Process analyzers play an important role in the efficient further processing of cracker...
LNG is natural gas in its liquid state with high energy density, which makes it useful for storage and transportation over long distances from the gas fields to the consumers. LNG is recovered from natural gas in large-scale liquefaction plants. Process automation is a key issue in LNG plants to ensure cost efficient plant operation and high product quality. Process analyzer deliver important data therefore.Siemens, a leader in process analytical instrumentation, has proven over decades its capability to plan,...
The ChArMEx (Chemistry and Aerosols Mediterranean Experiments) SOP2 (special observation period 2) field campaign took place from 15 July to 5 August 2013 in the western Mediterranean Basin at Ersa, a remote site in Cape Corse. During the campaign more than 80 volatile organic compounds (VOCs), including oxygenated species, were measured by different online and offline techniques. At the same time, an exhaustive description of the chemical composition of fine aerosols was performed with an aerosol...