Laser Induced Breakdown Spectroscopy or LIBS is an analytical technique that's been used in laboratories for many years. Most handheld LIBS analysers are used for fast sorting of alloys in scrap yards, and alloy identification and analysis in various applications within the metals industry.
During LIBS analysis, a focused pulsed laser hits the sample removing a very small amount of material from the surface. The sample is hit by thousands of pulses during a typical one-second measurement. The material is heated up to and exceeding 10,000 degrees Celsius. The temperature is so high that the atoms actually break up and form a plasma.
Despite these high temperatures, the sample does not get hot to the touch during analysis, and so it can be held in your hand safely during measurement.
What is happening within the atoms?
When the high power pulsed laser hits the sample, the electrons in the outer atoms’ outer shells are ejected. Because electrons in the outer shells are shielded by the inner shell electrons, they aren’t strongly drawn by the nucleus.
This means less energy is needed to eject electrons from the outer shells.
The ejected electrons create a vacancy, making the atom unstable. When the pulse stops, the plasma starts to cool down and the vacancy is filled by electrons cascading down from the outer electron shells. The excess energy released when the electrons move between two energy levels or shells is emitted in the form of element specific light.
For a typical metallic sample containing iron, manganese, chromium, nickel, vanadium, and so on, each element emits many wavelengths leading to a spectrum of up to thousands of peaks.
The wavelengths of light are collected through a fibre optic cable and then processed by the spectrometer.
Each element is associated with a specific spectral peak. LIBS spectra are quite complex with potentially hundreds or even thousands of lines for each element.
The concentration of the element is calculated from the intensity of the spectrum peak. Then an advanced algorithm is used to identify the sample type and calculate the concentrations.
Why do people choose the handheld LIBS technique?
It’s one of the fastest analytical technologies available today to identify and analyse metal alloys.
It takes just one second to measure virtually any alloy including aluminium alloys. That`s up to 20 times faster than handheld XRF.
A LIBS analyser is very robust because the detector is typically protected by sapphire glass, one of the hardest materials known today. So you can safely measure sharp and pointy objects, including turnings and shavings.
It’s also very easy to use. Just point, shoot and read the result from the screen.
The LIBS technique is virtually non-destructive. Following a LIBS analysis, the tiny burn spot left is so small it can be difficult to see with the bare eye.
From a regulatory and licensing point of view, much less is required from the user than, say, for XRF. Typically, no expensive licenses or time-consuming training classes are needed. However, it’s important to check your local legislation concerning issues such as the use of safety goggles. We strongly recommend using safety goggles when operating a Class 3B laser device.
LIBS is an excellent alloy identification tool which can also deliver a good chemistry when required.
Trace metals are found at low concentration in the ambient environment and are associated with adverse health effects.The gold standard metal measurement is Inductively Coupled Plasma Atomic Emission Spectroscopy (ICP-AES); however, sample preparation is laborious and expensive.The Xact 625 metals monitor uses non-destructive X-ray fluorescence technology. A built-in sample collection system enables near-real-time trace metal measurements in ambient air. Modifying the Xact monitor to also characterise metal...
This study converted sewage sludge into a carbonaceous catalyst via pyrolysis and employed it in the ozonation of hydroquinone. The catalyst was characterized by Mössbauer spectroscopy, X-ray photoelectron spectroscopy, temperature programmed desorption, scanning electron microscopy, transmission electron microscopy, Fourier transform infrared spectroscopy and X-ray diffraction. Intermediate products were detected by gas chromatography–mass spectrometry, and a pathway for hydroquinone degradation was proposed....
Oxford Instruments and das-nano demonstrate non-destructive wafer based thickness and resistivity metrology for PEALD TiN.
Commonly titanium nitride (TiN) thickness and resistivity wafer fab in-line metrology is based on ellipsometry and 4-point probe resistivity mapping. Alternative and relatively slower or more complex methods are X-ray photoelectron spectroscopy (XPS), X-ray reflectivity (XRR) and X-ray fluorescence (XRFS). TiN thin films are highly conductive and lose transparency for thicker layers which...
Polydopamine-modified granule organo-attapulgite adsorbent (PDA-GOAT) was facilely prepared via a dip-coating approach. The samples were characterized by scanning electron microscopy, X-ray photoelectron spectroscopy and Fourier transform infrared spectroscopy. Surface area and pore size were calculated from the Brunauer–Emmett–Teller method by N2 adsorption–desorption isotherm. The adsorption behaviour of methylene blue (MB) onto PDA-GOAT was systematically investigated. The experimental data revealed that the...
Nanoscale zero-valent iron (NZVI) was first assembled on magnetic carbon/Fe3O4 (CM) with a combination of hydrothermal and liquid phase reduction methods. The novel NZVI@CM magnetic nanocomposites have the merits of large surface area, unique magnetic property, low cost and environmental friendliness. They can be used for Pb(II) removal in aqueous solution. The materials were characterized by using transmission electron microscopy (TEM), X-ray diffraction (XRD), Fourier transform-infrared spectroscopy (FT-IR),...