Recalibration & Repairs
Visit www.apogeeinstruments.com/recalibration-and-repair/ for RMA information.
Model and serial numbers are found on a white label on sensor cables, on the back-side of meters, or on the underside of spectroradiometers.
Please send all RMA sensors back in their original, unmodified condition:
- please clean the sensor’s exterior and cord
- do not modify the sensors or wires, including splicing, cutting wire leads, etc.
- if a connector has been attached to the cable end, please include the mating connector – otherwise the sensor connector will be removed in order to complete the repair/recalibration
To determine if recalibratiion is necessary for pyranometers and quantum sesnsors visit our Clear Sky Claculator at www.clearskycalculator.com or download the mobile app.