Non-Aqueous and Aqueous Conductivity - Monitoring and Testing - Laboratory Equipment
Non-polar liquids such as toluene, hexane, dodecane, etc, are important for many novel products like electronic paper, batteries, and fuel cells. Dispersions prepared on their basis depend on ionic composition, which can be characterized with conductivity measurement. Utilization of this simple method can be accomplished only with a special conductivity probe capable of measuring conductivities of liquids over a very wide dynamic range, from 10-11 S/m up to 10-4 S/m.
There are two models that can conduct such measurement: stand alone Non-aqueous conductivity meter DT-700 and Option OP 0041 for DT 1202. Precision of a single measurement is ±(1%+10E-11S/m) over the complete range, and can be statistically improved for multiple measurements. This Figure demonstrate results of such conductivity measurements for several liquids with distinctively different conductivities:
Application of this instrument for characterizing non-polar liquids and dispersions can be found in several published papers, see for example:
Dukhin, A.S. and Parlia, S. “Dukhin, A.S. and Parlia, S. “Ion-pair conductivity theory fitting measured data for various alcohol-toluene mixtures across entire concentration range”, J. of the Electrochemical Society, 162 (4) H1-H8 (2015).
One of the most interesting application of this method is characterization of surfactant adsorption from non-polar liquids into porous materials. This application is described in the following paper:
Dukhin, A.S. and Parlia, S. “Adsorption of surfactant from non-polar liquid into porous material characterized using conductivity measurement”, Journal of Colloid and Interface Science 531, 523–532 (2018).